Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||15 August 2004|
|PDF File Size:||11.4 Mb|
|ePub File Size:||12.35 Mb|
|Price:||Free* [*Free Regsitration Required]|
Message 6 of 6.
Message 4 of 6. However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.
I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
Thanks for your help guys. When I gave hl4191a a try, though, it worked fine. I have not worked with that instrument, but the Agilent web site has the operating and service manual here.
If not, what would creating a driver for it entail? In particular, after setting up and initiating the instrument, I am initiating the manuak and reading it as follows:. Message 2 of 6. Would you mind posting the code you used on the A? You really need the manual.
Message 1 of 6. Message 3 of 6. Good luck with it Has anyone done anything on interfacing with the A since Matt C’s last post?
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Sorry, that last line of code there is supposed to read: I think the would probably take the old commands, but the ALSO has new commands. I suspect that this difference might be causing my problems. Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
Auto-suggest helps you quickly narrow down your manuual results by suggesting possible matches as you type.
Agilent / HP 4191A RF Impedance Analyzer
The forums emoticons took over for a bit there. I ,anual worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. Should I be looking at doing it with LabView or something else? In the code above, MHz is an arbitrary frequency.
Message 5 of 6. It’s 84MB, so I hope you have a fast link. Accepted by topic author Matt C. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: IIRC, the sweep parameters were something like this: Guess you never know until you try! Most Active Software Boards: