The methodology for measuring DQE in a digital detector has been standardized in IEC This standard applies to 2-D detectors used for general. ABSTRACT. The international standard IEC defines the measurement procedure for determination of the detective quantum efficiency (DQE) of digital. Measurement of the detective quantum efficiency in digital detectors consistent with the IEC standard: practical considerations regarding the choice of.
|Published (Last):||8 February 2006|
|PDF File Size:||9.5 Mb|
|ePub File Size:||12.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
This Part is restricted to digital X-ray imaging devices that are used for mammographic imaging such as but not exclusively, CR systems, direct and indirect flat panel detector based systems, scanning systems CCD based or photon-counting. If the document is revised or amended, you will be notified by email.
We have no document history for this standard. You can download and open this file to your own computer but DRM prevents opening this 26220 on another computer, including a networked server. This low frequency mottle significantly overestimates the NPS at low frequencies, which in turn will cause an under-estimation of the low frequency DQE. Please first verify your email before subscribing to alerts.
There was a problem providing the content you requested
As part of a larger evaluation we attempted to measure the detective quantum efficiency DQE of an amorphous silicon flat-panel detector using the method described in the International Electrotechnical Commission standard published in October You may delete a document from your Alert Profile at any time.
Qualitative evaluation of flat field images acquired with of ultra-high purity aluminum filtration An evaluation of ultra-high purity aluminum purchased from a second source Need more than one copy?
Qualitative evaluation of flat field images acquired at 71 kV RQA5 beam quality with 21 mm of ultrahigh purity aluminum filtration demonstrated a low frequency mottle that was reproducible and was not observed when the measurement was repeated at 74 kV RQA5 beam quality with 21 mm of lower-purity aluminum A subsequent evaluation of ultrahigh purity aluminum, purchased from a second source, suggests, that reduced grain size can be achieved by the process of annealing.
Add to Alert PDF. Standards Subsctiption may be the perfect solution.
We purchased scientific-grade ultra-high purity aluminum We have no amendments or corrections for this standard. Characteristics of digital X-ray imaging devices – Part 1: However, images acquired with this sample demonstrated striated defects attributed to the manufacturing method, which would preclude its use in radiographic imaging applications.
We will discuss problems encountered using ultra-high purity aluminum filtration for DQE evaluations. We recommend that the IEC and committees revisit their recommendation of using aluminum with a minimum purity of Our observations suggest that the non-uniformity of filter materials should be evaluated when measuring the DQE. All imaging tests were performed on a prototype amorphous silicon digital detector using the RQA5 beam quality.
Standards Subscriptions from ANSI provides a money-saving, multi-user solution for accessing standards. Problems with the Required Aluminum Filtration.
IEC | IEC Webstore
Already Subscribed to this document. Your Alert Profile lists the documents that will be monitored.
This finding was ultimately attributed to the larger grain size approximately mm of high purity aluminum metal, which is a well-known characteristic, particularly in thicknesses greater than 1 mm.
6220 standard is also available to be included in Standards Subscriptions. Please first log in with a verified email before subscribing to alerts.
Our observations of markedly visible nonuniformities in images acquired with high purity aluminum filtration suggest that the uniformity of filter materials should be carefully 622200 and taken into consideration when measuring the DQE. This part of IEC is not applicable to – 6220 X-ray imaging devices intended to be used in general radiography or in dental radiography; – computed tomography; and – devices for dynamic imaging where series of images are acquired, as in fluoroscopic or cardiac imaging.
The intended users of this part of IEC are manufacturers and well equipped test laboratories.
Subscription pricing is determined by: Determination of the detective quantum efficiency, specifying how the DQE should be measured and computed. This finding was ultimately attributed to the larger grain size of ultra-high purity aluminum metal.
Proceed to Checkout Continue Shopping. To achieve the radiographic beam conditions specified in the standard, we purchased scientific-grade ultrahigh purity aluminum This standard is not included in any packages. As the voice of the U.