View and Download ABB Advance Optima Uras 14 service manual online. Advance Optima Uras 14 Measuring Instruments pdf manual download. Manuals and User Guides for ABB Advance Optima Uras We have 1 ABB Advance Optima Uras 14 manual available for free PDF download: Service Manual. Spare Parts for ABB & Siemens Gas Analyzer & Sample Handling Systems is available. Home» Products» ABB Gas Analyzer Spare Parts» Module Uras
|Published (Last):||24 August 2013|
|PDF File Size:||11.21 Mb|
|ePub File Size:||20.32 Mb|
|Price:||Free* [*Free Regsitration Required]|
Structure The board is connected to the IR module circuit board. Temperature control board terminal strip X2 Remove the thermal link from the terminal strip and measure the thermal link continuity.
Uras 14 Gas Analyzer AO ABB AO *Refurbished* | eBay
Temperature detectors Temperature detector The temperature detector control T-Regl. Ask a question – opens in a new window or tab.
Page Temperature detectors, continued Temperature Temperature detector 1 is used for temperature compensation of IR components. Electrical connections What electrical The following connections are always present: Page 20 Summary, continued Figure Analyzer urax Uras 14 open 1 Calibration unit 2 2 Measurement cell 2 3 Main frame with aperture control 4 Beam path 2 aperture adjustment screw 5 Modulator with emitters not visible 6 Calibration unit 1 7 Infrared detector 1 8 Beam path 1 aperture adjustment screw 9 Measurement cell Continued on next page Chapter 7: IR detectors The example here is for the measurement of CO, NO, SO and O Uras detector configuration Note that while the oxygen sensor is not part of the Uras module it must always be configured together with the analyzer module.
Housing purge 1 sample gas path 1 beam path no gas module Hood with supplemental heater Location in module The hood encloses the optical components. Pump configuration Help Chapter 7: Hardware configuration, continued Detector Make sure the proper closing disk is installed.
In this chapter you will find the following information: Ask a question about this product Call Table of contents Page Chapter 1: Calibration unit Accessories Multimeter, PC with test- and calibration software The upper, lower and lateral travel paths of the calibration cell slide must be free of Visual check interference by cables and tubes.
Add to Watch list. Pressure Sensor Circuit Board Pressure sensor circuit board Location in module Depending on the analyzer module configuration, the pressure sensor is mounted in various positions on the main frame.
Emitter insert electrical passages Where? The module consists of a completely self-contained optical unit with the following elements: Mouse over to zoom – Click to enlarge. IR detectors, continued Pressure correction Continued urs next page Chapter 7: Open the large door on the analyzer housing.
Except for the following exceptions, use the black closing disk part number as a cap behind the applicable end IR detector in beam path 1 and 2. Solenoids General To calibrate the analyzer module from one to three optional solenoids can be installed and then set up using the Gas module menu item.
Abb Advance Optima Uras 14 Manuals
Connecting cable, continued Figure Connection cable 1 pin layout Chapter 3: Thermal link Accessories Multimeter, screwdriver Resistance Electrical value: Depending on the application, there can be from 1 to 4 detectors installed. Cross-sensitivity alignment, continued Execution Result correction The result of the CS correction is the CS factor for the corrected component.
The filter cell consists of a gold-plated tube. Between yellow and green approx. Structure The heated hood controls optical element temperature together with the heater Function installed on the main frame.
ABB Advance Optima Uras 14 Service Manual
Hold the cell up to a light and look through it. This tubing is suitable for most measurement applications.
Learn More – opens in a new window or tab Any international postage is paid in part to Pitney Bowes Inc. Page 36 Modulator, continued Figure Modulator Aperture wheel 1 Emitter 2 2 Aperture, beam path for reference chamber bab Emitter 1 4 Aperture, beam path for sample chamber Continued on next page Chapter 3: Perform manual zero and span calibrations with test gases.