View and Download YOKOGAWA EJXA installation manual online. EJX Series and EJA-E Series Differential Pressure and Pressure Transmitters. EJXA. Manual. Yokogawa Electric Corporation. EJX Series. HART Communication Type .. This manual describes the HART protocol communica- 1 EJXA Get Yokogawa EJXA Manual. Get all Yokogawa manuals!.
|Published (Last):||3 April 2009|
|PDF File Size:||20.76 Mb|
|ePub File Size:||2.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
The zero point adjustment can be The local push button on the integral indicator must made with a resolution of 0.
A non-hardening silicone group sealant is recommended. Use proper tools for all operations. Not applicable for output signal code F. The procedure for conducting these tests is as follows: Normally, only non- conductive pollution occurs. Capsule temperature sensor problem.
Yokogawa EJX110A User Manual
Page 13 Note 4. Page 4 iii 8. Maximum Input Voltage Ui: Comments to this Manuals Your Name.
Page 55 Revision Information: Either of these terminals may be used. Please do not ground the manuual on the RTD side of the cable. The following sorts of symptoms indicate that transmitter may not be operating properly.
Problems occurred during the warranty period shall basically be repaired free of charge. Page 71 Unit: Page 22 3.
Tighten the bolts securely and uniformly to a torque shown below, and verify that there are no pressure leaks. Do not pull the cable or subject it to excessive mechanical shock. Use a slotted screwdriver to turn the zeroadjustment screw. Follow the domestic electrical requirements RTD, Pt Model 8.
Wire cables through a flameproof packing adapter, or use Two cable glands are attached. To pipe by fastening a U-bolt to its mounting bracket. Page 80 Unit: Model And Specifications Check Installation of an Explosion- ej1x10a following precautions when selecting the installation Protected Instrument location.
Yokogawa EJXA : Manual
Check settings and change them as OV. This cap must for the process fluid gas, liquid, or steam. Maintenance of the transmitter is easy due to its modular construction. Since this will cause errors in the measurement output, select the proper piping method covered with a plastic cap to keep out dust.
Special Conditions for Safe Use locations: Page of 57 Go.
Yokogawa EJXA User Manual | 85 pages | Also for: EJXA, EJXA
NOTE This manual describes the hardware configurations of the transmitters listed in below. Don’t have an account? If you require further assistance, please contact the nearest Yokogawa office. The not be used in a hazardous area. Apply a permanent mark in the check box of the selected approval type on the certification label on the transmitter to distinguish ejx10a from unused approval types.
Connections of External Wiring to Terminal Box This cap must be removed before connecting the piping.