View and Download Hach Sc E instrument manual online. Analysis System. Sc E Measuring Instruments pdf manual download. Hach Sc Controller User Manual – Download as PDF File .pdf), Text File .txt) or read online. Hach Sc Controller User Manual . Hach Company, – .. 37 W with 25 W sensor/network card load 24 VDCpowered sc model: 24 VDC.
|Published (Last):||15 August 2013|
|PDF File Size:||18.10 Mb|
|ePub File Size:||5.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
Relay loads must be resistive. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. DANGER Do not replace any component unless power to all electrical circuits to the Controller have been switched off and the area is known to be non-hazardous. Specifying The Displayed Language Operation bach.
The turbidimeter sensor must be manuql within six feet of the controller unless an extension cable is used. The following two options are available: Page 46 Warranty, liability and complaints 7. Periodic inspection of these devices is recommended to check for degradation. Power terminal Electrical Installation 16 2. Wire gauge less than 18 AWG is not recommended.
Do not touch the new bulb with bare hands. Don’t see a manual you are looking for? The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value.
Troubleshooting Section 7 Troubleshooting Rinse dc100 deionized water before calibrating. Mounting bracket, controller, Cat. Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair.
Table of Contents Add to majual manuals Add. Page 33 Operation manaul. Supplied mounting hardware is shown in Figure 6, Figure 7, and Figure 8. Page 38 Operation 4. The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk.
Sensor Setup Menu Operation 4. If a service contract is taken out within 6 months of purchase, the warranty period is extended to 60 months.
HACH SC E INSTRUMENT MANUAL Pdf Download.
For the features available for each individual digital network port, refer to the nanual sheet supplied with the selected network card. System Startup Two types of verifications are offered. Drain body and clean the body and bubble trap.
The warranty period for instruments is 24 months. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure.
Setting Up System Security Operation 4. Replacing The Lamp Assembly Do not connect or disconnect electrical components janual circuits to the equipment unless power has been switched off jach the area is known to be non-hazardous. Table of Contents Section 1 Specifications Restore sample flow to the instrument.
Page 32 Operation 4. All configuration settings and calibrations can be changed.
Hach-Lange SC : Owner manual
Controller Dimension Illustrations Installation 3. Pay attention to all danger and caution statements. Push in and turn to secure the connection. The display will flash on startup, when a sensor error has occurred, when the hold outputs function has been activated, and when a sensor is being calibrated. Depends on sample Clean the sensor characteristics. Pull the lamp leads and connector through the hole in the lamp housing.
Warranty, liability and complaints 7. In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
The non-sensor end is connected to the controller as described in section 3. Don’t have an account? By pressing ‘print’ button you will print only current page. Relays K1, K2, and K3.
Hach sc100 1720E Instrument Manual
Tell us about it. Dry Verification System Startup Two types of verifications are offered. Enter text from picture: