The HP A is capable of making highly accurate measurements. In order to .. you should use manual range to get to the lowest range possible for the. Hp A Pdf User Manuals. View online or download Hp A User Manual. View and Download HP A user manual online. A Multimeter pdf manual download.
|Published (Last):||27 September 2014|
|PDF File Size:||10.62 Mb|
|ePub File Size:||4.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
To Use Reading Memory Disclaimer Redistribution and use of this article or any images or files referenced in it, in source and binary forms, with or without modification, are permitted provided that the following conditions are met: Loading Errors dc Volts HP A after cover removed.
Parity Selection rs Removed remnants, cleaned and repaired PCB using copper foil and installed fresh capacitor of same size and spec. SCPI The status system records various instrument conditions in three register groups: To Rack Mount The Multimeter Direct coupled to the ,anual and shunt.
Chapter 6 Application Programs Application Programs These filters allow you to trade-off low frequency accuracy for faster reading speed. Remote Interface Configuration HP A local 15V zener regulator.
Common Mode Errors Only the last six characters are manuql from the front panel, but all eight characters are required. This cable has a 9-pin female connector on each end. Chapter 3 Features and Functions Triggering Trigger Delay You can insert a delay between the trigger signal and each sample that follows.
HP 34401A Manuals
Chapter 3 Features and Functions Triggering Automatic Trigger Delays If you do not specify a trigger delay, the multimeter selects an automatic delay for you. Display Control Chapter 3 Features and Functions System-Related Operations Display Control To speed up your measurement rate, or for security reasons, you may want ho34401a turn off the front-panel display.
Status Reporting Commands Resolution Chapter 3 Features and Functions Measurement Configuration Resolution Resolution is expressed in terms of number of digits the multimeter can measure or display. Interpreting Multimeter Specifications Chapter 3 Features and Functions Calibration Overview To Secure Against Calibration You can secure the multimeter against calibration either from the front panel or remote interface. Reading Hold Chapter 3 Features and Functions Triggering Reading Hold The reading hold feature allows you to capture and hold a stable reading on the front-panel display.
HP 34401A User Manual
Errors In High Resistance Measurements Shift registers on AC circuitry and few other opamps were replaced as well, but that was actually unnecessary. The rate is set RS to baud when the multimeter is shipped from the factory. Chapter 3 Features and Functions Measurement Configuration Autozero When autozero is enabled defaultthe multimeter internally disconnects the input signal following each measurement, and takes a zero reading. LM is socketed, just like more recent xA instruments.
That is likely due to absence of fan or vent holes anywhere in the chassis. The security code may contain up to 12 alphanumeric characters. Rs Interface Commands Meter does power on, VFD screen lit up and keypad working normally. Check to see if you have enclosed the character string in hp34410a or double quotes and that the string contains valid characters.
In this article initial diagnostics, repair attempt and calibration will be covered. IEEE The multimeter must be in the wait-for-trigger state. Integration time affects the measurement resolution for better resolution, use a longer integration timeand measurement speed for faster measurements, use a shorter integration time.
It then subtracts the zero reading from the preceding reading. Rs Interface Configuration Measurement Configuration Chapter 3 Features and Functions Measurement Configuration Measurement Configuration This section contains information to help you configure the multimeter for making measurements.
| HP A repair and calibration worklog
To print the manual completely, please, download it. Autozero Chapter mqnual Features and Functions Measurement Configuration Autozero When autozero is enabled defaultthe multimeter internally disconnects the input signal following each measurement, and takes a zero reading.
Multimeter is in remote mode remote interface. This may be useful in applications where you want to allow the input to settle before taking a reading, or for pacing a burst of readings.