Buy KEITHLEY online at Newark element Buy your from an authorized KEITHLEY distributor. The Model Electrometer combines flexible interfacing capabilities with current sensitivity, charge measurement capabilities, resolution, and speed that are. Product Features: Programmable electrometer; Less than 1fA noise; >TW input impedance on voltage measurements; Charge measurements from 10fC to .
|Published (Last):||12 October 2018|
|PDF File Size:||17.34 Mb|
|ePub File Size:||18.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
The instrument also allows active cancellation of its input voltage and current offsets, either manually via the front panel controls or over the bus with IEEE commands.
The builds on the features and capabilities of the Keithley electrometers that preceded it. Model Image In a similar manner, offset currents can also keith,ey cancelled.
A similar circuit used also as a I-to-V converter, to allow electrometer to measure currents down with 10 A femtoamperes or 0. If you find typo or have any questions feel free to leave comment or contact us. Electrometers provide current measurement with lower terminal voltage than is possible when making DMM measurements.
Model front face Image 2: The Keithley Programmable Electrometer combines flexible interfacing capabilities with current sensitivity, charge measurement capabilities, resolution, and speed that are equal or superior to our earlier electrometers.
The offers the flexibility and sensitivity needed for a wide array of experiments, providing better data far faster than older electrometer designs.
Teardown Removing cover which is same as ModelC is easy task, just few screws and we are in. Total man-hours spent on this project around 1 hoursgive or take few. Model Image 8: A, May Model Datasheet Brochure As with all recent Keithley gear, schematics or detailed construction information is not available.
DMM Offset Currents Typically, offset currents in DMMs are tens or hundreds of picoamps, which severely limits their low current measuring capabilities compared to the with 3fA input bias current.
Was it worth doing? View Cart My Account Low Voltage Burden The ‘s feedback ammeter design minimizes voltage offsets in the input circuitry, which can affect current measurement accuracy. Data Sheet KB.
Disclaimer Redistribution and use of this article or any files referenced in it, in source and binary forms, with or without modification, are permitted provided that the following conditions are met: Built-in IEEE and RS interfaces make it possible to program all instrument functions over the bus through a computer controller.
Removing cover which is same as ModelC is easy task, just few screws and we are in. Power supply transformer is TR Rev.
Performance test … Conclusion …. These instruments usually have special low-leakage shielded input connectors, often with triaxial design with PTFE insulators.
| Keithley Electrometer repair
They usually have voltage, current, resistance and charge measurement functionality, just like usual DMMbut can do these measurements with minimal loading on the circuit under test. Electrometers are special ,eithley instruments, which key feature is very high input impedance. Bulk capacitors are listed in table below: Model Power supply transformer is TR Rev.
Any error signals that remain are negligible in comparison to those that can occur when measuring current with a DMM. Credits and thanks to ToddAlan and all keithlej readers to make this all possible.
Providing the finest test equipment solutions since 1992
In a feedback ammeter, the primary “voltage burden” is the amplifier offset voltage. Model front face. The instrument’s sensitivity makes it easy to determine the leakage resistance on capacitances up to 10nF or even on higher capacitances when a series resistor is used.
A normal-mode rejection ratio NMRR of 60dB allows making accurate low current measurements, even in the presence of line frequency induced currents, which is a common concern in production floor environments.
These instruments usually have special low-leakage shielded input connectors, often with triaxial design with PTFE insulators, Image 1: