Electromagnetic flow converter for combination with OPTIFLUX and WATERFLUX flow sensors. KROHNE 03/ – – TD IFC R05 en. The documentation is only complete when used in combination with the relevant documentation for the . 2. The high-performance solution. IFC is the only electromagnetic flow converter with 3 x diagnostics of instrument, application and .
|Published (Last):||13 February 2012|
|PDF File Size:||4.72 Mb|
|ePub File Size:||19.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
F hardware detection Existing hardware cannot be identified. Status Output And Limit Switches Avoid continuous direct sunlight. Kfohne measurement performance is excellent even in more difficult applications like mediums with low conductivity or mediums with high solid content or entrained air, corrosive and or abrasive mediums.
Start-up Status Table in Sect. S uncertain measurement Device maintenance necessary; measured values only conditionally usable. Each time a housing cover is opened, the thread should be cleaned and greased.
B3 information B3 information – B3.
Our Products Vissers Sales Corp. Number of Employees More than People. Look at the device nameplate to ensure that the device is delivered according to your order.
Measuring pipe not filled, function dependent on Fct. Calculation of mass flow with constant density of product; range: I control input A act. Only available if “manual” is set in C5. We draw your attention to the fact that data transmission over the Internet e. Warnings And Symbols Used This warning refers to the immediate danger when using this device in a hazardous atmosphere.
Table of Contents Add to my manuals Add. Display of the actual flow fraction for partly filled pipes. Any work done on the electrical components of the measuring device may only be carried out by properly trained specialists. Aluminium and aluminium sheet, partially polyester-coated Option Versions C ifd F: Display when setting parameters, 4 lines Current menu, sub-menu or function Number relating to 1 Denotes factory setting Denotes permissible value range Permissible value range for numeric values Currently set value, unit or function when selected, appears with white text, blue background This is where the data is changed.
Defective, replace electronic unit. Get Best Price Request a quote. F factory settings Error during the CRC check of the factory settings.
Krohne IFC 300 Flow Converter Manual PDF
If the device is not used according to the operating conditions refer to chapter “Technical datathe intended protection could be affected. This offers great time and cost benefits ic regard to procurement, engineering, operation and servicing. The electronics plug-in module is identical, regardless of the housing version, providing the same functionality and a standardized operating concept. Summary of Jfc for Krohne ifc Page 2: F sensor local Data error in the local data of the measuring sensor electronic equipment.
Reference value for on-site calibration; range: Settings Set in increments of 0. Set Free Units Make setting, see Fct.
Deposits in measuring tube or electrode short-circuit to ground. If status message still does not disappear, replace electronic unit. Assembly materials and tools are not part of the delivery. Krohne ifc Handbook 59 pages signal converter for electromagnetic flowmeter. These warnings must be observed without fail.
Page 14 – 1. A dialogue appears in the display. An external power supply is necessary to operate activation the subsequent devices Uext. To Reset Counters Either the connection is interrupted or the measuring sensor is not powered up.
The signal cables A type DS with double shield and B type BTS with triple shield ensure proper transmission of measured values.
Krohne ifc Manuals
Page 11 – Type, length and preparation of field cu If this does not happen within the specified time period, 30 device can be operated using the optical keys again.
If you require data that is more relevant to your specific application, please contact us or your local representative. The operating point of the 4 optical keys is located directly behind the glass pane.
A signal converter is used to amplify the signal voltage, filter it and convert it into signals for totalising, recording and output 3300. Krohne is a world-leading manufacturer and supplier of solutions in industrial process instrumentation. Spare Parts Availability Grounding Of Sensors primary Heads Warning: